Aquí puede consultar accesorios sobre SmartLipo.
Componentes y accesorios, piezas de mano y herramientas aportan versatilidad y eficiencia a nuestras estaciones de trabajo. Explore aquí la lista de accesorios para su dispositivo DEKA.
The Smartlipo reusable handpiece is made up of a handpiece with a fibre fixing device and a cannula that has to be linked to the distal part of the handpiece which is necessary to deliver the fibre into the body.
There are 2 types of available cannula:
– one with an external diameter of 1 mm that is used with 300 µm fibres (single use or reusable) and,
– one with an external diameter of 1,4 mm that is used with 600 µm fibres (single use or reusable).
Each type of cannula is produced in 5 different lengths in order to fit different treatment area dimensions. Cannula lenghts are: 10 mm, 15 mm, 20 mm, 25 mm, 30 mm.
Disposable kit including handpiece, telescopic cannula and fibre were developed and patented by DEKA to give doctors an instrument which does not need sterilization before the operation. The kit is…
DEKA provides 300 µm single use fibres for Dentistry. They are suggested for endodontics, parodontology, surgery and perimplantite treatments.
All fibres are made in silica and have 2 plastic protective tubes: a Teflon tube that starts from the laser connector and covers the fibre for the first 1,5 m; and a silicon tube which protect the remaining part of the fibre. The Teflon tube is stronger and is used to prevent shocks while silicon is more flexible and enhances the freedom of movement.
Fibres come in a double sterile envelope, ready to be used.
DEKA provides reusable fibres for Dentistry which can be sterilized in autoclave up to 10 times. Reusable fibres are 300 µm of diameter and they are suggested for endodontics, parodontology, surgery and perimplantite treatments.
All fibres are made in silica and have 2 plastic protective tubes: a Teflon tube that starts from the laser connector and covers the fibre for the first 1,5 m; and a silicon tube which protects the remaining part of the fibre. The Teflon tube is stronger and is used to prevent shocks while silicon is more flexible and enhances the freedom of movement.
Reusable fibres also have sterilization counters close to the connector. There are ten rings with numbers from 1 to 10. Each ring has to be removed before sterilization. Fibres come in a double sterile envelope, ready to be used.
A sterilization process or a prolonged use, tends to turn opaque fibres silica, especially in free end surface. Opacity can be seen by the reduction of the aiming beam brightness. The main effect of opacity is the reduction of fibre efficiency in delivering laser beam power. In order to regenerate fibre efficiency, DEKA provides a special knife to cut fibre end and create a new transparent fibre end surface.
Another useful tool is the fibre cleaver that is able to cut the silicon protective tube avoiding any damage to the fibre lateral surface.
After Laserlipolisi®, especially in big areas like abdomen or legs, it is possible to aspirate the lysate created with the laser (laser assisted liposuction). For this operation DEKA provides two cannulas for aspiration: one 15 cm long with 2 holes and another with 22 cm long with 3 holes. Aspiration cannulas are provided by DEKA in a NON sterile envelope. Doctors must sterilize them before use.
NOTE: Cannulas have to be linked to a pump for fat aspiration which DEKA do not provide.
Clicking on the link below you can download a pdf file with the summary table, equipped with all accessories available for the system, and relative codes.
¿Necesita más información acerca de nuestros productos o eventos?
Error: No se encontró ningún formulario de contacto.